Patient care delay.
The issue involves functionality in cerner millennium powerchart office and powerchart core and affects users that utilize the powerchart inbox and message center inbox.
In results to endorse or sign and review, if the user clicks ok and next multiple times in quick succession while attempting to sign a result or a document, the display could lag behind the system's processing of the action, and multiple results or documents could be signed without the user's review.
In message center, when clicking ok and next or accept and next, or when deleting or completing messages and moving to the next task, a document could be signed or a message could be deleted without the user's review.
Results could be endorsed or documents could be signed without physician review, which could impact patient care.
Cerner received communication that a patient's follow-up care was delayed as a result of this issue.