||February 17, 2007
||Terminated on June 06, 2008
|Recall Event ID
Micro Test Rolls -
||Anaerotest for microbiology
Specification: Performance Test Anerobic: white within 4-6 hours; aerobic: blue with 20 minutes
64271 Darmstadt, Germany
1.19034.0002 17.08.06 1 unit
UPC 4 022536 667928
||Item number 1.19034.0002
Lot # OC553145
EMD Chemicals Inc
480 S Democrat Rd
|For Additional Information Contact
|Lot # OC553145 of Anaerotest failed to remain in specification. QC indicator shows an anaerobic environment even if oxygen is present.
||A Recall Letter was issued by EMD Chemicals on 8/17/2006 to discontinue product usage and for product return.
|Quantity in Commerce
||Product was only sent to one US customer: Becton Dickinson, Cayey. Pureto Rico.
1 For details about termination of a recall see Code of Federal Regulations (CFR) Title 21 §7.55
2 Per FDA policy, recall cause determinations are subject to modification up to the point of termination of the recall.