||November 19, 2012
||Terminated on March 04, 2013
|Recall Event ID
Multi-analyte controls, all kinds (assayed) -
||Thermo Scientific MAS Omni IMMUNE and Omni IMMUNE PRO;
Liquid Assayed Integrated Immunoassay Control.
Intended for use as an assayed control for monitoring assay conditions in many clinical laboratory determinations. Assay values are provided for the specific systems listed. The user can compare observations with expected ranges as a means of assuring consistent performance or reagent and instrument.
||MAS Onmi IMMUNE:
Catalog Number/lot number/ expiration date:
OIM-101 - OIM14031 - 31 MAR 2014;
OIM-202 - OIM14032 - 31 MAR 2014;
OIM-303 - OIM14033 - 31 MAR 2014;
OIM-SP - OIM1403S - 31 MAR 2014.
MAS Omni IMMUNE PRO:
OPRO-101 - OPRO14031 - 31 MAR 2014;
OPRO-201 - OPRO14032 - 31 MAR 2014'
OPRO-303 - OPRO14033 - 31 MAR 2014;
OPRO-SP - OPRO1403S - 31 MAR 2014.
44660 Osgood Rd
|For Additional Information Contact
510-979-5000 Ext. 31814
|In house testing confirmed that CK-MS and Insulin claimed in both Omni IMMUNE and Omni IMMUNE PRO products are displaying reduced stability.
||Microgenics Corporation sent a Recall Notification letter dated August 20, 2012, to all affected customers. The letter identified the product the problem and the action needed to be taken by the customer.
Customers were instructed that immediate attention is required and identify the two affected constituents. The letter contains a list of the affected product with their catalog number, lot number and expiration date and request that the CK-MB and Insulin use be discontinued. The remaining components of the product may be used.
Evaluate quality controls according to your laboratory procedures.
Retain a copy of the letter for your laboratory records. If the product was forwarded to another laboratory, provide them with a copy of the letter.
Complete the attached Recall Response form and return within 5 days.
Questions may be directed to 1-800-232-3342.
|Quantity in Commerce
||2076 total in US , 596 total Foreign
||Worldwide Distribution--USA (nationwide) and the countries of Germany, Canada and Australia.
|Total Product Life Cycle
TPLC Device Report
1 For details about termination of a recall see Code of Federal Regulations (CFR) Title 21 §7.55
2 Per FDA policy, recall cause determinations are subject to modification up to the point of termination of the recall.
510(K)s with Product Code = JJY and Original Applicant = MICROGENICS CORP.