|Date Initiated by Firm
||November 26, 2012
|Date Posting Updated
||January 16, 2013
||Terminated 3 on March 07, 2014
|Recall Event ID
||Ninhydrin and l-leucyl-l-alanine (fluorimetric), phenylalanine - Product Code JNB
||NEOGRAM AMINO ACIDS AND ACYLCARNITINES TANDEM
MASS SPECTROMETRY KIT
Product Number: 3026-0030.
This kit is intended for the measurement and evaluation of amino acid, free carnitine, and acylcarnitine concentrations from newborn heel prick blood samples dried on filter paper. (Table 1 of Attachment 2) details the analytes measured by the kit. Quantitative analysis of amino acids, free carnitine and acylcarnitines and their relationship with each other is intended to provide analyte concentration profiles that may aid in the screening of newborns for one or more of several metabolic disorders. This kit is to be used for in vitro diagnostic use only, by trained, qualified laboratory personnel.
||Lot Numbers: 617341, 619236
940 Winter Street
Waltham MA 02451
|Truncated V-Bottomed Microplate Wells used in assays contain defective wells.
||Perkin Elmer initiated the recall on November 23, 2012. The Recall Letter will be delivered to the end customer by email facilitated through our PerkinElmer Technical Support group located in Akron Ohio.
Technical Support will be asked to make sure the customers receive the Recall Letter and implement the recommended precautions:
- Inspect the wells of row A and row H of the affected microplate lots for possible holes and cracks prior to use.
- If a hole or crack is visually detected in a well of the microplate, the well should not be used for testing.
- If a hole or crack is not visually detected in a well of the plate, it is required during use of the well, the volume of solution in the well should be visually confirmed to be consistent before and after incubation.
Please contact PerkinEimer Technical Support for further information (800) 321 -9632, select prompt 2.
|Quantity in Commerce
||180 kits (20 plates/kit)
|Total Product Life Cycle
TPLC Device Report
1 A record in this database is created when a firm initiates a correction or removal action. The record is updated if the FDA identifies a violation and classifies the action as a recall, and it is updated for a final time when the recall is terminated. Learn more about medical device recalls.
2 Per FDA policy, recall cause determinations are subject to modification up to the point of termination of the recall.
3 For details about termination of a recall see Code of Federal Regulations (CFR) Title 21 §7.55.
||510(K)s with Product Code = JNB and Original Applicant = PERKINELMER LIFE SCIENCES