||August 13, 2013
||StealthStation S7 Framelink Software version 5.4.
Aid for locating anatomical structures and planning surgical trajectories in open and percutaneous procedures.
||Lot number(s): Not applicable; All kits containing the Framelink Software version 5.4 are affected by this corrective action.
|Medtronic Navigation, Inc.
826 Coal Creek Cir
Louisville, Colorado 80027-9710
|For Additional Information Contact
|Medtronic Navigation, Inc. has initiated a field correction to their Framelink software version 5.4 which can be used on Medtronic Navigation StealthStation 57 and i7 systems or a Medtronic Planning Station due to the potential of discrepancy between the exam image and the displayed measurement or overlay under certain conditions of use.
||A Field Safety Notice was sent on 7/15/13 to users informing them that the Framelink software version 5.4 has been revised to correct the software anomalies described in this letter. A new version will be installed on their system(s) by a Medtronic Navigation representative. Until the software upgrade has been installed on their system, do not use high resolution examinations (greater than 256 x 256). If users are not sure of the exam resolution being used, firm recommend contacting the hospital imaging department or local Medtronic Navigation representative for assistance. Medtronic representatives will visit all identified hospitals and upgrade all systems to version 5.4.1. This will account for all consignees.
|Quantity in Commerce