|Note: this medical device record is a supplement. The device description may have changed. Be sure to look at the original PMA to get an up-to-date view of this device.|
|Trade Name||SYNVISC(HYLAN G-F 20)|
|Classification Name||acid, hyaluronic, intraarticular|
|Supplement Type|| Special Supplement|
|Supplement Reason|| labeling change - minor|
|Expedited Review Granted?|| No|
|Approval Order Statement |
Approval for the addition of the following statement to the synvisc physician package insert: "precaution: do not over tighten or apply excessive leverage when attaching the needle or removing the needle guard, as this may break the tip of the syringe. ".