|Note: this medical device record is a supplement. The device description may have changed. Be sure to look at the original PMA to get an up-to-date view of this device.|
|Trade Name||CONCERTO II CRT-D, CONSULTA DF4 ICD, CONSULTA ICD, MAXIMO II CRT-D, PROTECTA CRT-D, PROTECTA XT CRT-D|
|Generic Name||cardiac resynchronization therapy|
|Supplement Type||30-day notice|
|Supplement Reason|| process change: manufacturing|
|Expedited Review Granted?|| No|
|Approval Order Statement |
Additional inspection activity during the manufacturing of batteries.