|Note: this medical device record is a supplement. The device description may have changed. Be sure to look at the original PMA to get an up-to-date view of this device.|
|Trade Name||NOVOSTE BETA-CATH 3.5 SYSTEM|
|Classification Name||intravascular radiation delivery system|
|Supplement Type||30-day notice|
|Supplement Reason|| process change: manufacturing|
|Expedited Review Granted?|| No|
|Approval Order Statement |
Alternate method of manufacture of the beta-cath 3. 5f system flushing adapter.