|Note: this medical device record is a supplement. The device description may have changed. Be sure to look at the original PMA to get an up-to-date view of this device.|
|Trade Name||VIGOR(R) DR/SR PACEMAKER SYSTEM|
|Supplement Type||normal 180 day track|
|Supplement Reason|| change design/components/specifications/material|
|Expedited Review Granted?|| No|
|Approval Order Statement |
Approval fort he modification of the diagnostic (self-test) function of the mdoel 2909 multiple application utility (mau) to accommodate variability in watch dog timer (wdt) time-out periods.