|Note: this medical device record is a supplement. The device description may have changed. Be sure to look at the original PMA to get an up-to-date view of this device.|
|Trade Name||CERVISTA HPV HR|
|Classification Name||kit, dna detection, human papillomavirus|
|Supplement Type||30-day notice|
|Supplement Reason|| process change: manufacturing|
|Expedited Review Granted?|| No|
|Approval Order Statement |
Changes in the fill volume of the wash buffer and an increase in sample size of the incoming components being tested.