|Note: this medical device record is a supplement. The device description may have changed. Be sure to look at the original PMA to get an up-to-date view of this device.|
|Trade Name||VERIFLEX (LIBERTE) CORONARY STENT SYSTEM|
|Classification Name||stent, coronary|
|Generic Name||intravascular stent with delivery system|
|Supplement Type||30-day notice|
|Expedited Review Granted?|| No|
|Approval Order Statement |
Changes to the in-process endotoxin testing during component manufacturing.