LATITUDE® Paceart Integration Software, version 1.00 Model 6472.
LATITUDE Paceart Integration is software and therefore does not have any serial numbers. However, the software version affected is version 1.00 and is distributed on USB pen drives. These pen drives are lot number controlled and denoted as Model 6493.
When used in conjunction with the LATITUDE EMR Integration software transfers data files from the Boston Scientific LATITUDE Patient Management system to a clinic's Paceart system.
software verson 1.00
Recalling Firm/ Manufacturer
Boston Scientific CRM Corp 4100 Hamline Ave N Saint Paul, Minnesota 55112-5700
Contact the recalling firm for information
Manufacturer Reason for Recall
Boston Scientific has determined that under certain conditions, LATITUDE Paceart Integration (LPI) software
may not properly transfer data resulting in the incorrect summary of programmed parameter values in the ICD Summary tab display and Current Programming Tachy summary section of the Paceart report.
FDA Determined Cause 2
DESIGN: Software Design
Consignees were sent on 8/12/11 a Boston Scientific letter dated August 2011. The letter was addressed to "Dear Doctor". The letter described the product and problem. It also gave Background Information, Description of Potential Data Translation Error, and Additional Information and Action information.
A letter, including an HRS form, was also sent to all US LATITUDE Clinics that have active LPI software to notify them of the need to install the corrected LPI Software.
Quantity in Commerce
Nationwide Distribution -- AL, AK, AZ, AR, CA, CO, CT DE, DC, FL, GA, IL,IN, IA, KS, KY, LA,ME, MD, MA, MI, MN, MS, MO, MT, NE, NV, NH, NJ, NM, NY, NC, ND, OJ, OK, OR, PA, RI, SC, SD, TN, TX, UT, VT, VA, WA,WV, WI, and WY.