||August 01, 2013
||Siemens Luminos dRF Fluoroscopic X-Ray system
Fluoroscopic X-Ray system
||Model number 10094200, serial numbers 4024, 4077, 4043 and 4051
|Siemens Medical Solutions USA, Inc
51 Valley Stream Pkwy
Malvern, Pennsylvania 19355
|Siemens became aware of an unintended behavior when using the Luminos dRF with software version VD10A. The possibility exists that a software error may occur causing the table to tilt into an upright position and may collide with the floor. Four consignees were notified of this recall by Update Instruction XP067/12/S -- all four units have been corrected.
||Siemens initiated their correction of this product in May, 2013 by sending an Safety Update Instruction XP067/12S to resolve the software error to all affected customers. All units have been corrected.
Further questions please call (610)-219-6300.
|Quantity in Commerce
||US Distribution including the states of CA, IN, NJ and OK.