| ||Class 2 Recall|
Innova 4100 / 4100 IQ
||November 09, 2009
|Recall Event ID
System, Imaging, Pulsed Echo, Ultrasonic - Product Code IYO
||GE Healthcare, GE Innova 4100 / 4100 IQ. , ( The system consists of an a monoplane positioner, a vascular or cardiac table, an X-ray system and a digital detector)
|GE Medical Systems, LLC
3000 N Grandview Blvd
Waukesha, Wisconsin 53188-1615
|Potential pump failure associated with the Coolix 4000 chiller of the Innova single plane and biplane systems that may stop the chiller from working. If a chiller failure occurs, the user will be informed with a message on the n-room monitor stating that three minutes are left before the X-ray is inhibited. This could lead to a delay in treatment and possible additional X-ray exposure and contra
|DESIGN: Device Design
||Consignees were sent on May 1, 2009 a GE Healthcare " Urgent Medical Device Correction" letter dated April 16, 2009. The letter was addressed to Hospital Administrator/ Risk Manger, Mangers of Radiology/ Cardiology, and Radiologists/ Cardiologists. The letter described the problem, Safety Issue, Affected Product Details, Safety Instructions, Product Correction and Contact Information.
|Quantity in Commerce
||Worldwide distribution: USA, ARGENTINA, BELGIUM, BRAZIL, BULGARIA, CHINA, CROATIA, EGYPT, FRANCE, GERMANY, HONG KONG, HUNGARY, INDIA, ITALY, JAPAN, REPUBLIC OF KOREA, LATVIA, LIBYA, MEXICO, NORWAY, POLAND, RUSSIA, SINGAPORE, SPAIN, SWEDEN, TAIWAN, TURKEY, UNITED KINGDOM, ROMANIA, ISRAEL, UKRAINE, YEMEN, BELARUS, and AUSTRALIA.
|Total Product Life Cycle
TPLC Device Report
1 For details about termination of a recall see Code of Federal Regulations (CFR) Title 21 §7.55
2 Per FDA policy, recall cause determinations are subject to modification up to the point of termination of the recall.
510(K)s with Product Code = IYO and Original Applicant = GE HEALTHCARE
510(K)s with Product Code = IYO and Original Applicant = GE MEDICAL SYSTEMS, INC.
510(K)s with Product Code = IYO and Original Applicant = GE MEDICAL SYSTEMS, LLC