||March 18, 2011
||GE SENOGRAPHE ESSENTIAL FULL FIELD MAMMOGRAPHY SYSTEM (SENO ESSENTIAL)
The Senographe Essential system is intended to be used in the same clinical applications as traditional mammographic film/screen systems. It generates digital mammographic images which can be used for screening and diagnosis of breast cancer.
||All associated serial numbers.
|GE Healthcare, LLC
3000 N Grandview Blvd
Waukesha, Wisconsin 53188-1615
|During installation it was discovered that the duplicate of the 21 CFR 1010 Certification label for the X-ray Generator Cabinet was missing from the installation kits that allow installation of the Senographe Essential in a van (mobile unit).
||The firm plans to repair the defect or to bring product into compliance. The field action is being implemented to inspect the affected units and if found to be missing the certification label, a certification label will be applied. This will be carried out via a Field Modification Instruction planned to be released in March, 2011 with an estimated completion date of July, 2011.
|Quantity in Commerce