B. Braun Infusomat Space Infusion System: Class I Recall
MedSun: Newsletter #76, September 2012
FDA MedWatch Safety Alert
Recall due to the potential for breakage of the anti free flow clip catch located inside the infusion pump door. Breakage may occur when the IV set anti free flow clip catch is inserted improperly into the pump and the pump door is forced closed. Misloading of the anti free flow clip catch may create the potential for free flow of medication.
FDA MedWatch Safety Alert. B. Braun Infusomat Space Infusion System: Class I Recall. August 1, 2012.