Scientific Publications by FDA Staff
Biosens Bioelectron 2013 Jul 17;51C:1-7
Thousand-fold fluorescent signal amplification for mHealth diagnostics.
Balsam J, Rasooly R, Bruck HA, Rasooly A
The low sensitivity of Mobile Health (mHealth) optical detectors, such as those found on mobile phones, is a limiting factor for many mHealth clinical applications. To improve sensitivity, we have combined two approaches for optical signal amplification: (1) a computational approach based on an image stacking algorithm to decrease the image noise and enhance weak signals, and (2) an optical signal amplifier utilizing a capillary tube array. These approaches were used in a detection system which includes multi-wavelength LEDs capable of exciting many fluorophores in multiple wavelengths, a mobile phone or a webcam as a detector, and capillary tube array configured with 36 capillary tubes for signal enhancement. The capillary array enables a approximately 100x increase in signal sensitivity for fluorescein, reducing the limit of detection (LOD) for mobile phones and webcams from 1000nM to 10nM. Computational image stacking enables another approximately 10x increase in signal sensitivity, further reducing the LOD for webcam from 10nM to 1nM. To demonstrate the feasibility of the device for the detection of disease-related biomarkers, adenovirus DNA labeled with SYBR green or fluorescein was analyzed by both our capillary array and a commercial plate reader. The LOD for the capillary array was 5ug/mL, and that of the plate reader was 1ug/mL. Similar results were obtained using DNA stained with fluorescein. The combination of the two signal amplification approaches enables a approximately 1000x increase in LOD for the webcam platform. This brings it into the range of a conventional plate reader while using a smaller sample volume (10ul) than the plate reader requires (100ul). This suggests that such a device could be suitable for biosensing applications where up to 10 fold smaller sample sizes are needed. The simple optical configuration for mHealth described in this paper employing the combined capillary and image processing signal amplification is capable of measuring weak fluorescent signals without the need of dedicated laboratories. It has the potential to be used to increase sensitivity of other optically based mHealth technologies, and may increase mHealth's clinical utility, especially for telemedicine and for resource-poor settings and global health applications.
|Category: Journal Article|
|PubMed ID: #23928092||DOI: 10.1016/j.bios.2013.06.053|
|Includes FDA Authors from Scientific Area(s): Medical Devices|
|Entry Created: 2013-08-10||Entry Last Modified: 2013-12-20|