The patient required medical treatment for hyperkalemia and was started on convention hemodialysis.
Facility's biomedical technician inspected the machine and found the reason the machine did not pass the initial self-test was a problem with reposition of the pressure pod.
He ordered a new part.
"continuous renal replacement therapy machine transported.
Unable to prime the machine using filters from two different lot numbers.
Machine read "self test failure", "prime failure", and "00f0".
Patient had to receive medical treatment for hyperkalemia and conventional acute hemodialysis.