|Date Initiated by Firm
||July 19, 2012
|Date Posting Updated
||November 02, 2012
||Terminated 3 on June 20, 2013
|Recall Event ID
||Enzyme controls (assayed and unassayed) - Product Code JJT
||Cliniqa Liquid QCTM Cardiac Marker Control for i-STAT Level 1 (List Number 06F12-09; Lot Number 071039)
Cliniqa Liquid QCTM Cardiac Marker Control for i-STAT Level 2 (List Number 06F12-10; Lot Number 081039)
Cliniqa Liquid QCTM Cardiac Marker Control for i-STAT Level 3 (List Number 06F12-11; Lot Number 091039)
Usage: Assayed human plasma which may be used to verify the integrity of newly received i-STAT CK-MB cartridges.
||Cliniqa Liquid QC Cardiac Marker Control for i-STAT Level 1 (List #06F12-09; Lot Number 071039)
Cliniqa Liquid QC Cardiac Marker Control for i-STAT Level 2 (List #06F12-10; Lot Number 081039)
Cliniqa Liquid QC Cardiac Marker Control for i-STAT Level 3 (List #06F12-11; Lot Number 091039)
| Abbott Point Of Care Inc.
400 College Rd E
Princeton NJ 08540-6607
|For Additional Information Contact
||Mr. Joseph Kiceina
|Abbott Point of Care (APOC) has determined that for the Cliniqa Liquid QC Cardiac Marker Control for i-STAT Level 1 (List #06F12-09; Lot Number 071039); i-STAT Level 2 (List #06F12-10; Lot Number 081039); and i-STAT Level 3 (List #06F12-11; Lot Number 091039) results may be generated below the lower value assignment range for some vials.
||The firm, Abbott Point of Care, Inc. (APOC), sent an "URGENT RECALL NOTICE" dated July 2012 via FED-EX on July 19, 2012. The recall notice identified the product, problem and the action needed to be taken by the customer. The customers were instructed to discontinue use of the control lot numbers; return all unused controls on the enclosed Customer Reply Form; If another institution was provided the controls, provide a copy of this notice to them. Note: if any used vials gave acceptable product performance, a review of the cartridge performance and associated patient results is not necessary)
Note: APOC is shipping replacement products to customers facilities to minimize any disruption this might cause.
If you have any questions regarding this information, please contact Abbott Point of Care Technical Support at 800-366-8020, option 1 or your Abbott Point of Care representative.
|Quantity in Commerce
||154 (06F12-09 (74) boxes; 06F12-10 (35) boxes and 06F12-11 (45) boxes.) 1 box contains six, 3ml plastic bottles.
||Worldwide distribution: USA (nationwide) including states of: AZ, GA, ID, IL, LA, MD, MO, MN, NC, NV, OK, PA, TN, TX, VA and WI; and country of: Italy.
|Total Product Life Cycle
TPLC Device Report
1 A record in this database is created when a firm initiates a correction or removal action. The record is updated if the FDA identifies a violation and classifies the action as a recall, and it is updated for a final time when the recall is terminated. Learn more about medical device recalls.
2 Per FDA policy, recall cause determinations are subject to modification up to the point of termination of the recall.
3 For details about termination of a recall see Code of Federal Regulations (CFR) Title 21 §7.55.
||510(K)s with Product Code = JJT and Original Applicant = CLINIQA CORPORATION