|Date Initiated by Firm
||August 17, 2006
|Date Posting Updated
||February 17, 2007
||Terminated 3 on June 06, 2008
|Recall Event ID
||Micro Test Strips - Product Code J TX
50 test strips
Store dry and tightly closed.
Store at + 15 C to +25C
64271 Darmstadt, Germany
||Item # 1.15112.0001, Lot # OC553145
| EMD Chemicals Inc
480 S Democrat Rd
Gibbstown NJ 08027-1239
|For Additional Information Contact
|Lot # OC553145 of Anaerotest failed to remain in specification. QC indicator shows an anaerobic environment even if oxygen is present
||A Recall Letter was issued by EMD Chemicals on 8/17/2006 to discontinue product usage and for product return.
|Quantity in Commerce
||800 packages of 50 strips each.
||Product was only sent to one US customer: Becton Dickinson, Cayey. Pureto Rico.
1 A record in this database is created when a firm initiates a correction or removal action. The record is updated if the FDA identifies a violation and classifies the action as a recall, and it is updated for a final time when the recall is terminated. Learn more about medical device recalls.
2 Per FDA policy, recall cause determinations are subject to modification up to the point of termination of the recall.
3 For details about termination of a recall see Code of Federal Regulations (CFR) Title 21 §7.55.